Description of the goods or services required
CEH Lancaster require a laboratory water purification system number 2 to provide 300 litres per day of deionised water with 2 points of use (With scope to add ring main of up to additional 5 points at a later date).
Water demand: Total usage per hour 50 litres.
Water Quality: ISO3696 Grade 2
Operating Hours: 8hrs
Feed Water:Towns mains potable water supplied by United Utilities
Services: For the satisfactory operation of the system, the following services
(provisional): are to be provided by others, complete with connections and suitable means of isolation as required within 2 metres of the proposed equipment:
Feed water potable water supply as stated above, at 0.5m3 /hr, 3-6 bar under flowing conditions.
Drain at ffl with gully or tundish to accept waste water.
Electricity one off 400V 3 phase + n 50Hz fused isolator
Plant Room : For the satisfactory operation of the system, the plant room
Environment: environment needs to be controlled with frost protection and forced ventilation/air conditioning to maintain air temperature between 5oC and 30oC. Sufficient space should be allowed for access and egress for maintenance.
Primary water systems - 31161400
Collected and purified water - 41000000
Repair and maintenance services - 50000000
OFFER: CEH to be provided as part of the offer that the successful company will design, manufacture, inspect and test in works as appropriate, supply all equipment specified below, deliver to site, install, adjust, set to work, commission and handover with all relevant paperwork